Ad ID: 4470
Added: May 20, 2019
1.5 Torso x 2
1.5 Mark 9000 Shoulder coil
1.5 Classic Head Coil
1.5 4 CH CTL Coil
1.5 Quad Extremity coilGood
1.5 GP Flex Coil x 3
1.5 TMJ Coil
1.5 Invivo NV Array
1.5 MRI Devices NV Array
Magnet serial number R008 Manufactured April 1998
Coldhead recently replaced with refurbished date 10/2018
CSW 71 coldhead compressor Compressor static pressure below spec at 1.4psi
Helium lines and absorber also recently replaced. MM3 connected and configured for 4psi
Magnet has small amounts of ice and will not allow shim lead to be engaged. Ice will need to be removed before system can be ramped up.
Magnet and Covers were in good condition and Recently Painted Blue and White.
Magnet has been upgraded with Gradient cable Clamp kit and Gradient Lead Board.
PAC and SRI in Good Condition
There are 2 covers missing from laser lights.
Table in Good Condition up and down in out good.
Blower Box in good condition
Standard 3 Cabinet Excite
ACGD PDU manufactured July 2007
Excite 8ch Systems Cabinet manufactured July 2007
RF Cabinet with SSM and 8103 RF Amp manufactured July 2007
Gradient Chiller is Lytron manufactured 12/ 2007
There is no Coldhead Chiller included with scanner
Control Room and General System Info
System Computer is HP 8000
Keyboard and serial mouse present
Monitor in good condition.
Software Ver 11.1m4
There are no options MOD’S with equipment
Please contact us for Pricing.